The library is called Ex. To learn more about LabVIEW it is recommended that you go through the tutorial s and look over the material in the NI Developer Zone’s Learning Center which provides links to other materials and other tutorials. You can download updated drivers from the Instrument Driver Network. Message 3 of I used to Write my Query Command e.
|Date Added:||4 July 2014|
|File Size:||52.52 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Don’t forget to use the e3631s cluster to sequence things as this will tell you when you have finished reading or have got an error. Anyway, just substitute your drivers if different. The library is called Ex.
E3631A 80W Triple Output Power Supply, 6V, 5A & ±25V, 1A
As noted, just search for your model number. Both are available in 8. The attached vi shows proper use of error wires.
Message 10 of I prefered to writed my own driver using the GPIB commands. This can ypib with your code because you have the setting of the outputs in parallel. How i can avoid this jumping between read and write.
Providing the finest test equipment solutions since 1992
They basically perform the same function. Remember that reading usually involves two steps: Set and Read the voltage. Message 4 of Message 6 of Message 3 of You should get either the ” Plug and Play ” version of the ” Project-Style ” version depending on how you’re organizing your code.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. To learn more about LabVIEW it is recommended that you go through the tutorial s3631a and look over the material in the NI Developer Zone’s Learning Center which provides links to other materials and other tutorials.
However, you should not ignore the error cluster. You really should wire the error cluster through. Goib Active Software Boards: I got my drivers from the NI website. But not able set voltage and read the voltage from the PS in one program. You can also use the VISA resource wire.
All you code contains links to the drivers installed so anyone without the drivers installed cannot help you by looking at you code.
Keysight Power Supply, Triple Output, DC, Digital, EA
I’ve just moved company now and don’t have access tpib the software anymore, so you can’t have a copy – sorry. Force a sequential operation on all the VIs. Message 8 of Message 9 of Here is the way it is done with my drivers. Try looking at the manual on the agilent website http: I think the driver reads about bytes – you can check by looking at their code and make it yourself, but make it work for you and get it doing exactly what you want.
You must have a different set of drivers than I for the E Message 7 of Setting the volatge to three input4.