This is what I am thinking so far. I don’t know if you can setup such a “transistor”measurement!? Are you able to take a voltage sweep on a transistor? Nudge I like Keithley intsruments but they are going to obsolete themselves if they don’t snap to it and write drivers that work. Message 5 of Message 4 of I think that the plus with the conductance measurement software from Keithley are designed for resistor-like device:
|Date Added:||23 May 2004|
|File Size:||35.41 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 4 of Message 10 of Is this a VI that you had created? Message 2 of Unfortunaltely keithley engineers are strongly committed not to provide the block diagram.
Also, I would contact Keithley concerning the functionality of the and A. Message 3 of Can you tell me exactly what is “auto-test sequence running? I am trying to use the “differential conductance mode” in software “KI Are you able to take nanovoltmetdr voltage sweep on a transistor?
This may be the source of the warning; the sweep is starting and stopping at 0 A. In my case, I need to plot a keithlej graph which has source-drain voltage Xgate voltage Y and conductance Z.
I want to edit the program and make it convenient to me, as I suppose with my limited labview labbview it will take more time if I strat from scratch. It is possible that there is some setting that needs to be made on these particular devices to voltage sweep a resistor versus a transistor.
At this point, you may want to contact Keithley.
Keithley Nanovoltmeter – Instrument Driver – National Instruments
Does anyone has the block diagram for KIx. This is what I am thinking so far. As it says in the warning message, you may also want to check the connections of your transistor and verify that it is enough nanovoltmter to be measured. Message 7 of It is an example software written by the Keithley. This voltage sweep function works fine over a resistor, but not over a transistor.
Thanks for your labivew.
I don’t know if you can setup such a “transistor”measurement!? Since this is an executable that has been created by Keithley, there is no way that anyone other than Keithley can provide access to the code. They will be better able to answer any questions regarding signal connections with the and I think that the plus with the conductance measurement software from Keithley are designed for resistor-like device: Good luck in your applications! But it seems trying to locate a resistance value for the device under testthat is unable to accomplished on a transistor.
Message 6 of For some reason, I was unable to open the data sheets and manuals on Keithley’s website. Message 1 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
The Keithley driver 282a not a supported National Instruments driver, but I’ll try to help you the best I can. As a side note: Message 9 of Most Active Software Boards: