Press the and key, and set the number of times with the keys. Function Function Function Function No. Head B of OUT1 3: Refer to page Blow the dirt off with clean air or wipe it off. Is the power cable properly connected?
|Date Added:||9 November 2008|
|File Size:||37.69 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Page 80 3 Function Settings Stabilizing the Measurement by Filtering The following three types of function can be set for the filter.
What’s missing? Tell us about it.
Press the key to register the setting. Refer to “List of Default Values and Setting Ranges” page for the setting range and its restriction. The scaling of the analog current output 4 to 20 mA is ksb by calculating the electric current value to the voltage value by using the following formula.
Select the corresponding type from the following list. Page 29 1 Before Use Installing and Connecting the Heads and Optional Parts Installing the Head 1 Adjust the distance between the head and the measuring target, and fix the head with the screws at the two installation holes.
Page 5 RSC Command details This section describes the details of the incoming commands and the response commands that are sent when the keyyence commands are properly processed.
Press the key, and select function “k” with the key. Reference The display panel fixing screw does not drop from the display panel fixing case. Refer to for the layout of each terminal.
Press the key to return to the measurement state. Within the tolerance Red: Page 45 1 Before Use Setting Mode In the Setting mode, every time the key is pressed, the setting content advances in this order: Head A of OUT2 4: Page Specifications of the LK-G Series This chapter describes the specifications of the controller and the head, outside dimensions, and characteristics such as spot diameter. Function Settings Describes the functions and setting procedures of the Head settings, Output settings, Common settings, and Environment settings.
In addition, for ensuring the safe connection of the cord, leave 65 mm space or more in front of the terminal panel of the controller. Page 72 3 Function Settings Functions “A” and ” ” specify the following sub-functions. Page 94 3 Function Settings Outputting the Analog Output Without Holding The internal measurement value before being retained keyene outputted when the analog-through is set to ON while the measurement value is retained in the measurement mode.
Page 40 1 Before Use Combining the separate type lkk-gd500 1 1 Install the display panel by aligning it with the two protrusions on the display panel fixing case.
LK-GD – Separate type Display panel | LK-G series | KEYENCE International Belgium
Page 3 Function Settings Setting the Operations of the Equipment Environment Settings This section describes the operation settings of the equipment. When the measurement value is “— Remedy Check the wiring and connect properly. Panel attachment ring Two claws Fix the cable by fitting it along the guide. Function Function Function Function No. The Eco mode is activated in ksyence seconds after the last key operation during the measurement mode.
KEYENCE Lk-gd500 LKGD500 Laser Displacement Sensor Controller
Minimum display unit Unit Displayable range g-0 0. Failure to do so may cause malfunction.
Waveform by filtering with lk-fd500. The following functions can be specified by the settings. Be sure to read this section thoroughly before using the LK-G Series. The function of each measurement mode is as follows. Releases the holding when the measurement value is reset.
Page 3 Lj-gd500 Settings 5 6 3 7 Press the key, and specify the program No. Note These commands are accepted only when the controller is in the “Normal mode”.